[SQM] DL Retrieve-all speed-up
anthony at unihedron.com
Tue Feb 7 16:43:39 UTC 2017
Hello all DL users,
We're back on track now.
Software flow control is now implemented to resolve the fast binary
DL-Retrieve issue on slow computers. The change is available in:
- Firmware version 47
- UDM version 91
This has been tested on slow netbooks and the data came through intact
where it had failed before.
The firmware is included with the latest UDM:
On Fri, 3 Feb 2017 10:59:11 -0500, Anthony Tekatch <anthony at unihedron.com> wrote:
> Hello all DL users,
> It has come to my attention that the current binary transfer speed-up
> method misses some data when being retrieved from a slow computer like a
> slow netbook.
> The data set include records like this with invalid datestamps:
> If you have a slow computer, please continue to use the standard
> "Retrieve All" button for now.
> A solution is being investigated right now.
> Best regards,
> Anthony Tekatch
> On Sat, 28 Jan 2017 20:04:38 -0500, Anthony Tekatch
> <anthony at unihedron.com> wrote:
> > Hello All,
> > We have been able to radically reduce the data download time for the
> > SQM-LU-DL meter. A typical 30 minute download can now be done in
> > 65 seconds. Using the new "Retrieve All (bin)" button.
> > There are changes in the firmware and UDM software. The firmware
> > change (feature 45) is included with the latest UDM version (version
> > 85) and is fairly easy to install on your system without worry of
> > losing existing data or calibration settings.
> > You can get the latest UDM software for Windows from here:
> > http://unihedron.com/projects/darksky/cd/Windows/
> > Best regards,
> > Anthony Tekatch
> > Unihedron
> > http://unihedron.com
> sqm mailing list
> sqm at unihedron.com
More information about the sqm